The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Published (Last):||28 June 2006|
|PDF File Size:||20.70 Mb|
|ePub File Size:||11.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
Chapter 3 Features and Functions Amplitude Modulation AM Modulation Depth The modulation depth is expressed as a percentage and represents the extent of the amplitude variation. The default is Linear.
Select from 2 mHz to 20 kHz. Table of Contents Add to my manuals Add. Manula avoid multiple waveform changes, you can enable the burst mode after you have set up the other burst parameters. The default is Vpp.
The default is 10 Hz. Select whether the function generator uses true-high or true-low logic levels on the rear-panel Trig In connector for an externally-gated burst.
However, only one label for example, Freq or Period is displayed for each softkey at one time. The default is Sine. This connector accepts TTL-compatible levels and is referenced to chassis ground not floating ground.
33220A Function / Arbitrary Waveform Generator, 20 MHz
Manal Tutorial In order to achieve the best performance from the Agilent A, it may be helpful for you to gain a better understanding of the internal operations of the instrument.
At this point, the function generator outputs a continuous sweep from 50 Hz to 5 kHz if the output is enabled. Calibration error; setup is out of order Certain calibration setups must be performed in a specific sequence in order to be valid.
Press the key to view the list of available help topics. The default is VPP. The default is 1 ms. Chapter 5 Error Messages Manula Errors Settings conflict; selected arb is missing, changing selection to default If you delete an arbitrary waveform from non-volatile memory after storing the instrument state, the waveform data is lost and the function generator will not output the waveform when the state is recalled.
AUTO command on page for more information. Features and Functions This chapter makes it 32320a to look up all the details about a particular feature of the function generator. MIN selects the minimum duty cycle for the selected frequency and MAX selects the maximum duty cycle see restrictions below. The factory setting is enabled ON. Burst Mode Overview The following is an overview of the steps required to generate a burst. Press the Interp softkey to enable or disable linear interpolation between waveform points this feature is available from the front panel only.
Modulation Chapter 7 Tutorial Modulation Modulation Modulation is the process of modifying a high-frequency signal called the carrier signal with low-frequency information called the modulating signal. If you specify a low level that is greater than the existing high level, the function generator will automatically set the high level to be 1 mV greater than the low level. Press the key to enter the numeric entry mode. Single, External, or Internal Carrier Waveforms: The default is OFF.
In the triggered mode, the number of cycles specified by the burst count is output at the waveform frequency. TRIG command see belowa TTL-compatible manuaal waveform with the specified edge is output from the rear-panel Trig Out connector at the beginning of the sweep or burst. Chapter 4 Remote Interface Reference Simplified Programming Overview Simplified Programming Overview This section gives an overview of the basic techniques used to program the function generator over the remote interface.
This section describes the low-level commands used to program the function generator to output a pulse waveform.
The specified waveform data is divided into samples such that one waveform cycle exactly fills waveform memory see the illustration below for a sine wave. This section describes the low-level commands used to program the function generator. Front-Panel Number Entry You can enter numbers from the front-panel using 33220q of two methods.
The security code is stored in non-volatile memory. This section is only an overview and does not give all of the details you will need to write your own mamual programs.