BD CaliBRITE beads are designed for use with FACSComp or AutoCOMP software and the FACS family of flow cytometers (FACSCalibur, FACSort, FACScan. values for BD Calibrite beads. To edit, see page A Target file is also created for. HLA-B Although used by. BD FACSComp software, the file is not editable. Product Name: BD CALIBRITE BEADS. Synonyms: BD CALIBRITE BEADS; CALIBRITE BEADS. CAS: MF: MW: 0. EINECS: Mol File: Mol File. BD CALIBRITE .
|Published (Last):||22 September 2007|
|PDF File Size:||15.5 Mb|
|ePub File Size:||4.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Mix bead vials by gentle inversion or very gentle vortexing prior to use. BD Calibrite beads are used to determine the appropriate compensation settings.
Always refer to the appropriate application note or reagent IFU. All forms are provided in stabilized, buffered saline with 0. A minimum channel separation must be met for the scatter and fluorescence parameters. Gently mix the BD Calibrite bead vials, then add 1 drop of beads to each tube as indicated in the table below.
Wellington, Auckland, New Zealand bdbiosciences. Perform a Sensitivity Test using Tube B. Reagents are sufficient to perform 25 tests. NOTE Invert bead vials completely when adding a drop to the tube. For information on use, refer to the appropriate instrument manual.
BD Calibrite™ – BD Calibrite PerCP-Cy Beads – BD Biosciences
Use the same staining method and run in parallel with the test samples. BD Calibrite beads are for in vitro diagnostic use.
UV Bead lab with graph. Documents Flashcards Grammar checker. Observations of greater variations on a single instrument can be indicative of instrument instability.
I dont have a photo of any of my first Optimization and Quality Control Because leucocytes have different optical properties than BD Calibrite beads, optimization of instrument settings with cell samples is important.
See examples in Optimization and Quality Control on page 4. Optimize instrument settings following two-color setup using a blood sample stained with any combination of monoclonal antibodies that identifies separate non-overlapping cell populations, such as FITClabeled and PE-labeled monoclonal bexds.
BD Calibrite PerCP-Cy5.5 Beads
This allows cells to be distinguished from sample debris or background signal and for dimly stained cells to be distinguished from unstained cells. Because BD Calibrite beads simulate unstained cells and cells that have been stained labeled with fluorochrome-conjugated antibodies, the beads are used to adjust the instrument settings before cell samples are run on the flow cytometer. The channel separation and PMT voltages for each of the four parameters should be maintained in a daily log to track instrument performance.
It is not licensed for any other use. Corrective action might be required if the average separation varies by more than 2. Next, the software adjusts fluorescence compensation using a mixed-bead suspension containing equal amounts of the appropriate BD Calibrite beads.
Label two 12 x mm polystyrene tubes Tube A and Tube B. If deterioration is suspected, prepare a new bead suspension and check instrument conditions. Adjustment is similar for PerCP-Cy5.
BD CALIBRITE BEADS
If this occurs, manually adjust the settings. Refer to the information appropriate to the instrument setup you are performing.
The drop should be cloudy, indicating the beads are properly mixed. Figure 1 through Figure 4 show examples of optimization for two- three- and four-color applications.
Beads used beyond their stability begin to show a decrease in separation between unlabeled and labeled populations, possibly resulting in Sensitivity Test failure.
A thorough investigation of sucrose Optimizing Scatter Figure 1 shows a lysed whole blood LWB sample from a normal donor before and after optimization. How to make the beaded number line. Forward scatter FSC and side scatter SSC instrument sensitivity are measured by the mean channel separation between the light-scatter signal of the beads and background signal electronic and optical. After the instrument settings have been determined, BD Calibrite beads are used to evaluate instrument sensitivity.
Do not use after the expiration date shown on the label. This product is licensed for sale only for in vitro diagnostics. Preparation of Test Suspensions Prepare all bead suspensions immediately prior to use. Optimization following three- and four-color setup can vary depending on the application. One bottle is sufficient to perform 25 tests. Prepare a blood sample daily from a normal donor.
It might be necessary to adjust the FSC and SSC amplifiers so that all leucocyte populations are on scale, and to adjust compensation and threshold settings see Bead 1. Record PMT voltages and channel separations obtained for each parameter in a daily log sheet. See Xalibrite 1 through Figure 5 for examples. Both scatter and fluorescent light signals are detected. The 3-color kit contains these beads plus a PerCPlabeled bead.
Do not dilute PerCP-Cy5.